We are pleased to offer some of the most competitive pricing on quality, pre-owned aesthetic equipment available. We invite you to shop and compare our prices with our competitors. If you can find a price that's lower on a quality laser, we will make every attempt to meet or beat that price.
We understand the need for value in today's economy. We find great quality lasers at low prices and pass the savings onto you. All used devices come with a minimum 90 day limited warranty.
2007 Aerolase Lightpod Neo Nd Yag 1064nm; Excellent Cosmetic and Operating Condition, Very Low Pulse Count & Very Lightly Used, Single Owner; Recently Serviced June 2019; Reason for Selling: Closed Practice; Normal Cycle Pulse Count = 30,600, Super Cycle Pulse Count = 130,050; Includes: Handpiece, Collimated Lens, Focused Lens, Footswitch, Hard Carry Case, Operator Manual, Warning Sign, (2) Operator Eye Wear with Cases, Patient Eye Wear with Case and 90 Day Warranty.jp
Nd Yag 1064nm
Rise time, seconds
AC power source
110V/60Hz or 220V/50Hz
6.0 A max
Internal cooling system
14” x 19” x 7”
Model:MicroPulse 1064-XT 1.5
The single most versatile aesthetic and dermatology laser. Period.
The LightPod Neo is a New Laser Platform
Advances in technology tend to make products smaller and more affordable, yet more powerful at the same time. Aerolase has done exactly that in the field of aesthetic lasers. Internally air-cooled LightPod technology utilizes compact, high-power electronics, which replaces the traditional design concept of bulky, stationary water-cooled lasers, eliminating the internal water circulating system (the primary reason for the extreme size and weight of traditional lasers) as well as the fiber optic cables and lightguides – which are common repair trouble-points of this previous generation of lasers. Also, with a single handpiece design, there are no costly consumables such as tips or extra handpieces and includes an industry leading 1,000,000 pulse warranty.
Unique 650-microsecond Technology
The breakthrough of 650-microsecond pulse technology enables the production of high and low fluences ranging from 4 j/cm2 to 255 j/cm2 all within this single 650µsec pulse duration. Through clinical trials and research, Aerolase established this as the optimal pulse duration for enabling LightPod lasers to deliver energy deep into the targeted tissue with minimal absorption by epidermal melanin. This avoids thermal overstressing of the skin, eliminating treatment pain and diminishing the risk of any adverse effects commonly caused by the previous generation of long-pulse lasers. As a result, laser procedures can be performed without needing to cool the skin during treatment, even on skin of color. This, in turn, saves the practitioner a lot of cost, time and mess due to the elimination of cooling gels, sprays and anesthetics.
High-Powered 650-Microsecond Technology Overview
The LightPod Neo 1064nm laser’s ability to perform efficaciously without skin cooling is due to its ability to deliver the same range of laser energy density at the skin as traditional longer pulsed (i.e., 3 msec to 30 msec) Nd:YAG 1064 nm lasers, but in a high-powered, yet shorter 650-microsecond pulse duration. Although the melanin in the epidermis is exposed to the same laser fluence as it passes through to the deeper dermis layers, the time of this exposure is 5 to 50 times shorter when using the 650-microsecond pulse duration. This results in a lesser amount of energy absorption in the epidermal melanin. As a result, there is less heat generation within the epidermis, allowing treatment with no skin cooling and with virtually no discomfort. This also results in a higher percentage of laser energy reaching the targets including hair follicles in the deeper dermis, with the same fluence at the skin tissue, as compared to the traditional longer pulsed lasers. Furthermore, because the 650-microsecond pulse duration is shorter than the Thermal Relaxation Time of the skin tissue surrounding the follicle (estimated to be 800 microseconds), the laser-generated heat is effectively sealed within the follicle because it doesn’t have time to conduct outward to the surrounding skin. Therefore the temperature of the follicle and consequent tissue destruction are greater with the 650-microsecond pulse than with traditional (3 msec to 30 msec) pulsed 1064nm lasers.
* Shipping is included for the lower 48 United States only. Any shipment outside the lower 48 United States will be shipped at an extra cost. If you live outside this area, please contact us for a free shipping estimate.